Ask Question About Product
High-precision electric centrifugal sampler Uses:
Electric centrifugal sampler for a variety of seed, food, feed precision division. The electro centrifugal sampler can accurately divide samples in seconds, especially for small grain crops.
High-precision electric centrifugal sampler features:
Decimation speed, the motor driving the rotary feed hopper beneath the aluminum pan was centrifuged movement, rapid completion of the mixing process and the like uniformly
refined aesthetic appearance, chrome plated, rust and corrosion.
Adjustable speed, can adjust the appropriate speed according to different seeds.
It is especially suitable for small and medium-sized grain grains, which makes up for the inconvenience of using different types of instruments for different seeds of traditional dividers.
High-precision electrodynamic centrifugal sampler Technical parameters:
- - Sample amount: 200-1500g
- - Sample error: Less than ± 0.5% per kilogram
- - Dimensions: 300 × 300 × 550mm
- - Net weight: 18.6kg
- - Voltage: 220V 50HZ
High-precision electric centrifugal sampler use method:
- Place the instrument on the desktop first, adjust the screw at the bottom of the instrument, and tighten the screw after it is in the horizontal state.
- Refer to the level meter on the upper part of the instrument. The level of the instrument is related to the precision of the sample.
- Adjustment speed: according to different materials, the motor is first adjusted to the appropriate rotation speed (it can also be adjusted with the work during work).
- Check whether the instrument is in the off state. After confirming, turn on the power and then put the grain into the funnel. (Note: 1, the grain must be more than 1/2 of the funnel; 2. When loading the funnel, the seeds must always fall in the center), turn on the power, the instrument began to quickly sample. Complete within 5 seconds.
- For better accuracy, repeat it twice.